One of my old test gears waiting for my attention in the cellar was a HP 3562A Dynamic Signal Analyzer. Back then I bought this device mainly for doing audio analysis (I was building my HiFi system, mostly based on Nelson Pass' schematics), mainly to display spectrum responses and compute ...
This is the part 2 of the series about my HP 3562A Digital Signal Analyzer, quickly describing the Digital Section of the instrument.
The unit is quite capable, since the main CPU is a 68000 (not exactly of small CPU for the time).
The 8 boards are:
- A1 ...
This is the part 3 of the series about my HP 3562A Digital Signal Analyzer, quickly describing the Analog Section of the instrument.
The analog section consist in 6 boards:
- A30: Analog Source
- A31: Trigger
- A32, A34: Input ADCs
- A33, A35: Input
This board is mainly ...
This is the part 4 of the series about my HP 3562A Digital Signal Analyzer, quickly describing the HP 1345A Digital Display unit as well as the repair I had to do on it.
The display unit is a HP1345A Digital Display without the option 714 (wich adds a ...