dim. 18 septembre 2016
In order to build a replacement display for the HP34970A data acquisition unit, I needed to understand the communication protocol between the CPU board and the front panel assembly.
The unit is built with four boards:
- A1 is the main bard, with the PSU, the main controller and ...
sam. 17 septembre 2016
After a summer pause, I'm back on my HP34970A replair project. In the previous post in this series, I've started to reverse ingineer the serial protocol between the CPU board and the fonrt panel, and implement a prototype of replacement display.
I bought the bigest OLED ...
dim. 26 juin 2016
As explained in the previous post in this series, I've started to sniff the serial protocol between the main board and the display panel so I can replace the failed VFD by an OLED or TFT display managed by an arduino or similar.
The serial communication takes place ...
mer. 22 juin 2016
This is quick follow up of my previous blog post on my attempt to fix my recently acquired HP 34970A.
In my hurry to replace the failing FRAM (FM24C04) chips, I also replaced the FRAM chip on one of my 2 plugin modules.
Bad idea: the module then provoques an ...
lun. 13 juin 2016
I recently "scored" a HP 34970A with the DMM included and 2 plugin cards (HP 34901A and HP 34907A) for less than 200€. The plugins should be in working condition, but the main switch unit was almost "given" with the plugin modules since it is non functional.
It's in ...
mar. 19 avril 2016
After having mostly destroyed the PSU of my Tek 2445, I've decided to be a bit less foolhardy and I've subscribed to the yahoo Tek group. I discovered that such a failure has already been reported there, also after a full recap of the preregulator and regulator of ...
dim. 17 avril 2016
This post is a quick presentation and teardown of a small bench power supply I bought on ebay a few weeks ago for something like 40€.
It's an old unit, not sure it's exact age, but the components inside the unit seem to have date codes around 1971 ...
sam. 16 avril 2016
A few weeks ago, I purchased a cheap electronic load from AliExpress (this one to be precise). It's a ZHIYU model ZPB30A1. It's specifications are:
- constant current mode
- max voltage: 30V
- discharge current: 0.2A to 9.99A (+/- 0.7% + 1 digit)
- max dissipated power: 60W
- voltage measurement ...
dim. 27 mars 2016
After a quick repair of the PSU of my Tek 2445, I've started a more systematic replacement of the capacitors.
But I was also a bit worried because of a dirty noise coming from the preregulator area.
After having replaced most of the electrolytic capacitors, the dirty noise became ...
sam. 19 mars 2016
This is a quick follow-up of the first part of this series.
I've received a few parts from digikey, including a couple of 22µH Sumida CLS62 inductors (I bought two of them in case I f**k up one of them trying to solder it with my Weller WS50 ...